With the Indicator ON, press and hold the G/N/T/Menu button until MEnU is
displayed. When the G/N/T/Menu button is released, CAL is displayed.
Press the G/N/T/Menu button, SPAN Y is displayed.
Press the G/N/T/Menu button, -C- is displayed.
The scale base MUST be stable
during this period as it establishes a zero point.
After a few seconds, the requested weight value is displayed.
The sample illustration indicates a 5kg weight value (Cal Point CP was set for 5kg).
Place the indicated mass on the platform. Keep the platform stable during this period.
Press the G/N/T/Menu button, -C- is displayed while the Indicator stores the reading and then displays the weight of the mass.
If the calibration was successful, the calibration weight is displayed and the calibration data is saved automatically. If unsuccessful, refer to the troubleshooting section.
Remove calibration masses from the platform.
NOTE: If the Indicator is to be used for legal for trade or legally controlled applications, it must be calibrated and the jumper removed from the CAL connector in order to lock the menus. The top cover of the Indicator should be free to gain access to the CAL jumper. Refer to section 3.1 for sealing for legal for trade use.
David Ortega Peña